View and Download HP A calibration manual online. Synthesizer/Function Synthesizer HP A Operating And Service Manual. Frequency ( pages) . The Operating and Service Manual contains infor¬ mation required to install, operate, test, adjust, and serv¬ ice the Hewlett-Packard Model A. Find great deals for HP a Function Generator Operating and Service Manual . Shop with confidence on eBay!.
|Genre:||Health and Food|
|Published (Last):||6 May 2010|
|PDF File Size:||5.7 Mb|
|ePub File Size:||3.1 Mb|
|Price:||Free* [*Free Regsitration Required]|
Oscilloscope, with delayed sweep -hp- Model A a.
Oscilloscope -hp- Model A Function. High Stability Reference Option Sine Wave Amplitude Calibration. Instructions for programming Data Transfer Mode are included in Paragraph Change A amplitude to 3Vp-p, and remove the voltage divider from the circuit. Line 1 The sweep limits 3 to 10 are on the same range.
3325A Synthesizer/Function Generator
The dc offset is also checked in these tests. Sine Amplitude Control Path. Don’t show me this message again. Do NOT use a The sweep speed is determined by calculator and A data transfer and processing times.
Programming Execution Functions The V and mV keys enter peak-to-peak value of ac functions. The AC voltmeter reading should be Otherwise, both kits are available separately by their -hp- part numbers. See Figure A.
Set the A output as follows: The output level is a nominal 0 dBm sdrvice 50 ohms. Observe the spectrum analyzer for spurious responses. Seal the shipping container securely. If shipping to a Mamual office or service center, attach a tag indicating the type of service required, return address, model number, and full serial number. The lines should not be separated by more than ten divisions on the display.
Positive Slope Ramp Frequency.
HP A Service Manual download, schematics, eeprom, repair info for electronics experts
This bit can be monitored by the controller to determine when the A is ready for more data. This serrvice a Safety Class 1 instrument.
Service Group F – Fractional N This key is useful when an error is made w’hile entering data. Time shown is in addition to programming time. Adjust the oscilloscope controls to measure the high and low voltage levels of the sine square wave. Lighted indicator in the center of this key indicates High Voltage Output is on. The power jack and mating plug of the power cable meet International Electrotechnical Commission IEC safety standards.
Sine wave amplitude and flatness A Performance Test Record is located at the end of this section. Function Peak-to-Peak rms Max. Program data contains an error. The universal unlisten command removes all listeners from the bus, allowing only the listencr s designated by the device s listen parameter to receive information. The A responds to a Serial Poll, which is conducted in the following manner: This connector is Part No. Output amplitude is 1 mV to 10 V peak-to- peak.
Square Wave Rise Time and Abberations. Because the instrument turns on in Data Mode 1, Mode 2 must be programmed remotely.
It calls attention to procedure, practice, condition or the like, which is essential to highlight. This section contains the procedures required to adjust the A to meet its specifications in Table Enter the program into the calculator.
Programming Binary On or Off Functions.